Galleries | Nanomechanics
Contact Resonance Tunes on PZT
For domains with an antiparallel (180°) orientation, conventional PLLs drive the PFM frequency away from
resonance. (Top) Amplitude, red, and phase, blue, cantilever response over antiparallel domains. In the
measurement, phase is offset by 180° over anti-parallel domains (see curves on the right). (Bottom) PFM phase
signal driving the cantilever off resonance. Note the increased noise in the phase signal away from the
resonant frequency. This increased noise would be apparent in an image as well. Imaged with the MFP-3D AFM.